FLUKE 80T-IR PDF
80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
|Country:||Trinidad & Tobago|
|Published (Last):||4 December 2012|
|PDF File Size:||2.32 Mb|
|ePub File Size:||5.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Please enter a message. You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice. Reactivate by switching the UUT off and on again.
Fluke Infrared Temperature Probe M# 80t-ir
Thank you for your feedback. Sending feedback, please wait The product detailed below complies with the specifications published by RS Components. With the UUT powered on, check that the battery voltage is at least 7.
Replace the battery if necessary. Search the history of over billion web pages on the Internet.
Record the reading displayed on the DMM. Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure. Save this item to a new parts list.
Power up the UUT and allow it to stabilize for five minutes. The 80t-ri information relates to product sold on, or after, the date shown below. The product does not contain any of the restricted substances in concentrations and applications banned by the Directive, and 80t-lr components, the product is capable of being worked on at the higher temperatures required by lead—free soldering.
Set the DMM to the 3. Repeat steps 1 through 5 for the remaining verification points called out in Table 2.
FLUKE 80T IR 80T E Instruction
In between measurements 80t-irr the UUT to an environment not affected by the heat source. Save to an existing parts list Save to a new parts list. Save to parts list Save to parts list. Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody plate.
Fluke 80T-IR Infrared Temperature Probe
You have chosen to save the following item to a parts list:. If your instrument fails to pass any performance specifications, return it to the Eluke Eactory or Service Center for repair. Verification Points Model Blackbody Temperature at 0. Accuracy figures are valid for a period of one year.
Please select an existing parts list.
Fluke 80t-ir Temperature Probe 80TIR | eBay
The Manufacturers reserve the right to change this Information at any time without notice. The Manufacturers disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from your use of or your inability to use the Information downloaded from this website.
Adjust the setpoint of the Blackbody to output each of the temperatures in Table 2 at 0. Refer to Figure 1 for equipment configuration.
May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent accuracy, i.
Add to a parts list.